Heart device maker recalls implant after five deaths
USA Today, October 27, 2008
Medical device maker Thoratec Corp. has ordered the recall of certain batches of a small mechanical heart pump, saying five people have died while using the device. Thoratec urged patients to have their implants checked after confirming 27 reports of cases in which wear and fatigue to an electrical wire required the devices to be replaced, the company said in a statement. The reports occurred over five years of clinical experience with 1,972 implants. In five cases, the device could not be replaced and the patients died.
- CMS Sets 2014 Pay Rates for Hospital Outpatient and Physician Services
- FDA hopes hospitals will switch to newly regulated pharmacies
- The 5 Biggest Healthcare Finance Trouble Spots
- Not-for-Profit Hospitals Find Opportunity Amid Uncertainty
- The Most Polarizing Topics in Healthcare IT
- Nonprofit Hospital Outlook 'Negative' in 2014
- How CPOE Will Make Healthcare Smarter
- Why You Should Involve Patients in Nursing Handoffs
- Are ACOs Really Different from HMOs?
- Rise of the Chief Strategy Officer